Installing software for multiple wavelength meters or even two versions for the same instrument is possible.
Legacy software or instruments may experience compatibility issues. Please receive support from customer service in such cases.
Some optional features require separate hardware parts. Please make sure to assemble them before software installation!go on
The installation software is deployed on a flash drive. It‘s stored in the envelope that contains the manual and certificate.
Automatic installation, simply follow the prompts.
For advanced users desiring custom settings. Follow the prompts.
Some wavelength meters have an external power supply. Please connect the power supply before connecting the USB cable (included in shipment). Please avoid using USB cables longer than 3 meters – that might cause communication protocol difficulties with Windows.
Please note: Pay attention to the damage thresholds specified in the switch manual. A few µJ (µW @ 1 s exposure) or even less are usually enough to achieve a measurement. Take special care when working with pulsed lasers.
All wavelength meters feature FC/PC sockets.
Use the FC/PC:FC/PC patchcord
between wavemeter and switch and between
switch and laser.
Your measurement results will be impaired when you apply
FC/APC connectors to a wavelength meter input.
Simply run the desktop shortcut.
Depending on your light source adjust the software settings in the graphical user interface.
A) Select preferred unit
B) Select pulsed or continuous
C) Select Adjust exposure manually or select “Automatic”
D) Start the measurement
The Long-Term program is set to automatically start with the measurement. After completion of your measurement save the long-term data on your computer. You can reload at any later time or use the data with any other statistical program.
Don't hesitate to contact our service team for support.